IMO FTPC第5部分，ISO 5658
熱通量儀是一種水冷箔熱通量儀，範圍為0 – 5 W /cm²，並帶有校正證書。儀表的輸出以mV為單位，並且可以在合適的記錄設備（未提供）或可選的數據採集和軟件系統上進行監視。
電氣：230 Volts AC 50 Hz 或 115 Volts AC 60 Hz
氣體：設備的輻射板可以燃燒丙烷或天然氣。 典型的氣體供應要求是在200 mm水位計的供應壓力下流量最高為0.8升/秒。
尺寸：輻射板的整體尺寸約為480 mm x 280 mm
Spread Of Flame Apparatus
IMO FTPC PART 5, ISO 5658
The Spread of Flame Apparatus measures the flame spread of materials in marine and railway applications.
The IMO marine specification includes a collection hood for heat release measurement.
The Apparatus is constructed in accordance with the requirements of the Standard and consists essentially of two main items, the Radiant Panel Framework and the Specimen Support Framework, and a number of supplementary items.
Specfication and Features
Radiant Panel Framework
The support framework for the radiant panel is manufactured from all welded square tubular mild steel members, and includes all the associated pipework and control valves for the radiant panel. The radiant panel, which has overall dimensions of approximately 480 mm by 280 mm and consists of porous refractory tiles mounted in a stainless steel housing, is mounted on the support framework on a rotating steel ring, allowing for the testing of specimens in different orientations.
The radiant panel is designed for use with natural gas or propane gas and the gas is mixed with air prior to entry to the panel. Pressure governors and control and shut off valves are provided for both the gas and the air. Variable area flowmeters are incorporated in the supply pipework to allow the gas and air supplies to the radiant panel to be controlled and monitored.
A small control box is fixed to the framework which contains indicator lamps and push buttons associated with the gas safety system.
Specimen Support Framework
The framework which supports the test specimen is also made from all welded square tubular mild steel sections. One side of the framework contains steel guides which carry the specimen holder. Facility is available for location of a test specimen in the horizontal face upwards orientation.
When in the test position, the surface of the specimen at the end adjacent to the radiant panel is brought close to a test flame issuing from a stainless steel tube fixed to the support frame and supplied with a gas/air mixture from the main gas supplies on the radiant panel support framework.
For the IMO Specification a heat release stack is mounted on the specimen support framework and directly above the test specimen, such that the combustion products rising from a vertically mounted specimen are channelled into the stack.
The specimen support framework is linked to the support frame for the radiant panel by spacers and fixing bolts.
The stainless steel specimen holders are manufactured in accordance with the requirements of the Specification. A spring loaded quick action clamp for easy loading and retention of the test specimen is also provided.
This consists of a calibration board and a heat flux meter. The calibration board is manufactured from a calcium silicate based board material and is provided with a series of holes which, when the calibration board is inserted into the test apparatus, positions the heat flux meter at prescribed calibration locations.
The heat flux meter is a water-cooled-foil heat-flux meter with a range of 0 – 5 W/cm², and comes with a calibration certificate. The output from the meter is in mV and can be monitored on a suitable recording device (not supplied) or from the optional Data acquisition and software system.
Thermocouple Conditioning Device
This consists of a specially manufactured device which enables the outputs from the thermocouples on the heat release stack to be conditioned as required by the Specification.
Technical Data & Requirements
Electrical: 230 Volts AC 50 Hz or 115 Volts AC 60 Hz
Gas : The radiant panel of the apparatus can burn propane or natural gas. Typical gas supply requirements are a flow of up to 0.8 litres/second at a supply pressure of 200 mm water gauge.
Air : An air supply of up to 10 litres/second and at a supply pressure of approximately 300 mm water gauge is required
Dimensions: The radiant panel has overall dimensions of approximately 480 mm by 280 mm
|2019.11.07~11.11 9:30-17:00 2019台中自動化機械暨智慧製造展臺中國際展覽館(烏日區中山路3段1號),本公司於度量衡專區攤位559展出,歡迎各界先進蒞臨參觀指教！謝謝|
|Molecular Analysis 產品印度代理商|